This readme file contains additional information concerning the driver. A list of contact information for other countries is available on the Agilent Technologies website at http: After detailing a bit more the process do you still think there might be a connection drop out between the GPIB interface and HP? The SCPI commands look similar for both, but there are probably subtle differences that are the result of new features in the Agilent C. Did you try other commands which take a longer time to execute? Message 6 of
|Date Added:||6 January 2010|
|File Size:||39.2 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Did you get this working? I managedto get communication between the equipment HP and LabView.
It includes a comprehensive on-line help file which complements the instrument manual. It includes a “Function Panel”. After this I can see the cursor on the bottom of labbiew I V graph on the monitor of the sliding from left to right when settinp a Sweep of Voltage of SMU1 from 0 to 1 V, pts. Two elements of the standard do not apply to the instrument since it is not a VXI instrument.
Agilent Analyzer – IEEE (GPIB) Driver for LabVIEW – National Instruments
At the same time I get an error on the monitor of D00 Message 1 of 6. You must also provide an include path for hpb. The only exceptions are that it does not have a soft front panel or a knowledge base file. Thanks for the feedback. It’s alright, I now have a code that works perfectly! If i were you, i would contact KeySight formerly known as Agilent formerly known as HP what that message on the device is meaning.
Do you have labfiew experience with this? However, for maximum portability, the instrument lsbview should use uppercase characters only. Transportability across different computer platforms 2. This would suggest me that there is communication through the GPIB.
What exactly is stopping us from doing this? Message 4 of 6. It recognises the equipment by setting it to the default GPIB address That’s where your example comes from.
Agilent Technologies / Keysight Technologies 4156B
A higher level of support from Agilent Technologies for the compiled driver 3. It includes a Visual Basic include file.
Implementation of the addressing scheme is vendor specific and some vendors support mixed cases. I have a very old VI with a very old driver where you can obviously not choose an SMU or supply any source but can only read.
Message 3 of 6. Moreover I attach some printscreens where it can be seen how I defined the channel and source. Message 6 of The help file presents application programming examples, a cross-reference between instrument commands and driver functions, and detailed documentation of each function, with examples. But if you attach code I can test on a Message 2 of 6.
The driver uses Pascal calling conventions. I will try nevertheless labvies find a solution. I have done many kinds of sweeping on this one, but this is my first time with sampling and I can’t seem to figure that one out. I am not sure if the Instrument Driver for the B adds custom messages. You can specify lsbview a total sampling time or a specific number of 415b6.
Norbert Norbert CEO: